How do Ct values on the 24.192 GE IFC differ from values obtained on the Biomark 96.96 GE IFC?
Because of the difference in chamber size between the 2 IFCs, you can expect to see a shift in threshold cycle (Ct) values. The Ct values obtained from the 24.192 GE IFC are approximately 2 cycles higher than on the 96.96 GE IFC. You can add 2 cycles of Preamplification before running the 24.192 GE IFC to mitigate this effect.
For Research Use Only. Not for use in diagnostic procedures.